Comparison of the biometric measurements calculated with Zeiss IOL-Master and WaveLight OB-820
Received 25 January 2017
Accepted for publication 23 February 2017
Published 21 April 2017 Volume 2017:11 Pages 753—758
Checked for plagiarism Yes
Review by Single anonymous peer review
Peer reviewer comments 2
Editor who approved publication: Dr Scott Fraser
Georgios Labiris,1,2 Panagiota Ntonti,2 Konstantinos Ntitsos,2 Andreas Katsanos,2 Haris Sideroudi,2 Vasilios Kozobolis1,2
1Department of Ophthalmology, University Hospital of Alexandroupolis, 2Eye Institute of Thrace, Alexandroupolis, Greece
Background: The primary objective of this study was to investigate the level of agreement between IOL-Master and OB-820 ocular biometers.
Materials and methods: In this prospective randomized case series, we measured the anterior chamber depth (ACD), the axial length (AL), the corneal radii (R1, R2), the ratio of mean corneal radius and spherical equivalent and the corneal astigmatism (cylinder [Cyl]) before and after cataract extraction surgery.
Results: Significant differences between pre- and postoperative data were observed for ACD for both biometric devices (P<0.01) and Cyl parameter in IOL-Master. Range and 95% limits of agreement (LoA) were clinically significantly different for AL parameter pre- and postoperatively and for R2 and radius and spherical equivalent postoperatively (P<0.001). The rest of the parameters presented sufficient 95% LoA, which imply good agreement.
Conclusion: In clinical practice, the IOL-master and OB-820 should not be used interchangeably due to discrepancy in the important AL parameter. Both biometers may provide consistent results regarding Cyl, ACD and R1.
Keywords: cataract, IOL-Master, OB-820, biometry, level of agreement
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]