The SinuSonic: reducing nasal congestion with acoustic vibration and oscillating expiratory pressure
Received 14 April 2019
Accepted for publication 24 July 2019
Published 28 August 2019 Volume 2019:12 Pages 305—310
Checked for plagiarism Yes
Review by Single-blind
Peer reviewer comments 2
Editor who approved publication: Dr Scott Fraser
Alyssa Cairns,1 Richard Bogan1–4
1Sleepmed, Inc., Columbia, SC, USA; 2Bogan Sleep Consultants, LLC, Columbia, SC, USA; 3The University of South Carolina Medical School, Columbia, SC, USA; 4The Medical University of South Carolina, Charleston, SC, USA
Correspondence: Alyssa Cairns
SleepMed, Inc., 700 Gervais Street, Suite 200
, Columbia, SC 29201, USA
Tel +1 803 779 5454
Introduction: Nasal congestion is common, burdensome, and costly. Current treatments are limited by partial/temporary relief and untoward side-effects. The goal of this study was to evaluate the performance of a novel, non-pharmacologic nasal device designed to reduce nasal congestion via simultaneous administration of acoustic vibration and gentle oscillating expiratory pressure.
Materials and methods: Patients were recruited from a tertiary care sleep clinic and all reported moderate-to-severe nasal congestion for >2 weeks (N=14; 64% female; 71% Caucasian). Visual analog scale (VAS) (10 items) quantifying nasal congestion and ease-of-breathing were administered before and after SinuSonic application for 2–5 mins. Global and clinical impressions of change were assessed post-administration.
Results: Wilcoxon signed-rank tests indicated that post-test ranks were statistically improved from pre-test ranks for both VAS measures (congestion p=0.002; ease-of-breathing p=0.003). A binomial test indicated that the proportion of patients with ≥ minimal improvement on clinical and global impressions of change was higher than expected (100% vs expected 75%, p=0.018).
Conclusion: Overall, outcomes were encouraging from this small pilot study with effect sizes in the moderate to large range and no reports of discomfort. It is probable that this device will provide acute, and possibly chronic, relief of nasal congestion with minimal side-effects.
Keywords: humming, sinus, congestion
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]