Back to Journals » Medical Devices: Evidence and Research » Volume 2
The OxyMask™ development and performance in healthy volunteers
Authors Paul J, Hangan H, Hajgato J
Published 11 December 2008 Volume 2009:2 Pages 9—17
Review by Single anonymous peer review
Peer reviewer comments 1
James E Paul1, Horia Hangan2, Julius Hajgato3
1Department of Anesthesia, McMaster University, Hamilton, ON, Canada; 2Faculty of Engineering, The University of Western Ontario, London, ON, Canada; 3Southmedic Inc., Barrie, ON, Canada
Background: The OxyMask™ is a unique, open-style, oxygen mask that was originally developed in 2005. The original mask was modified, using computational fluid dynamics numerical simulations, with the goal of allowing it to produce a wider range of FiO2. This analysis was used to guide the modification of the mask shell and the location for the oxygen diffuser.
Methods: The new OxyMask was attached to 10 healthy subjects and used to deliver escalating levels of oxygen (1.5, 2, 2.5, 3, 5, 10, 15, 20, 25 and 30 LPM) for 90 seconds at each level and the resulting FiO2 was recorded (at the lips) from 5 consecutive measurements at each oxygen flow rate.
Results: Mean FiO2 was 25.4% at 1.5 LPM of oxygen, 30.1% at 2 LPM, 36.5% at 2.5 LPM, 41.8% at 3 LPM, 57.6% at 5 LPM, 74.4% at 10 LPM, and 80.1% at 15 LPM. Each FiO2 achieved at these escalating oxygen levels was significantly greater than all the previous levels. The mean FiO2 was 82.8 at 20 LPM, 84.2% at 25 LPM and 84.3% at 30 LPM. All of these values on average were not significantly greater than the FiO2 achieved with 15 LPM. In a few subjects a maximum FiO2 of 90% was reached.
Conclusion: The original OxyMask was successfully modified so that the second generation of the mask can provide a wide range of FiO2, from 25% to 90%, while keeping its unique open design.
Keywords: oxygen, oxygen masks, oxygen therapy, OxyMask™, OxyArm™, clinical trial, computational fluid dynamics (CFD), equipment design, biomedical engineering
© 2008 The Author(s). This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.