Retinal oximetry using ultrahigh-resolution optical coherence tomography
Authors Ye, Jiang H, Shen, Lam BL, Cabrera DeBuc D, Ge, Sehi, Wang J
Received 5 September 2012
Accepted for publication 1 November 2012
Published 12 December 2012 Volume 2012:6 Pages 2085—2092
Checked for plagiarism Yes
Review by Single-blind
Peer reviewer comments 6
Yufeng Ye,1,2 Hong Jiang,2,3 Meixiao Shen,4 Byron L Lam,2 Delia Cabrera DeBuc,2 Lili Ge,2,4 Mitra Sehi,2 Jianhua Wang2
1Hangzhou First People’s Hospital, Hangzhou, China; 2Bascom Palmer Eye Institute, 3Neurology, University of Miami, Miami, FL, USA; 4School of Ophthalmology and Optometry, Wenzhou Medical College, Wenzhou, Zhejiang, China
Background: The purpose of this study was to investigate the repeatability of retinal oximetry using slit-lamp adapted ultrahigh-resolution optical coherence tomography (SL-UHR-OCT).
Methods: SL-UHR-OCT was developed and fringe patterns were obtained for a major retinal artery and a major retinal vein. A-scans at the central wavelengths of 805 nm and 855 nm were analyzed for calculating optical density ratios (ODRs), from which the percentage oxygen saturation was calculated. Measurements were made on two occasions for each person. Repeatability and coefficients of repeatability were calculated.
Results: The mean ODRs of the artery were 0.79 ± 0.86 and 0.88 ± 0.97 in sessions 1 and 2, respectively. The mean ODRs of the vein were −0.08 ± 0.69 and 0.14 ± 0.77 between the two sessions, and were significantly lower than that of the artery (P < 0.05). The coefficients of repeatability were 1.44 and 1.81 for the artery and vein, respectively. The mean oxygen saturation of the major retinal artery was 94% ± 45% and 98% ± 51% in sessions 1 and 2, respectively, and the mean oxygen saturation of the major retinal vein was 48% ± 36% and 60% ± 40% between sessions.
Conclusion: Optical coherence tomographic oximetry for evaluating retinal oxygen saturation was subject to variation, although the averaged measurements in repeated sessions were matched. Further work on reducing variation will be needed.
Keywords: ultrahigh-resolution optical coherence tomography, oximetry
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]