Optimizing the detection of methicillin-resistant Staphylococcus aureus with elevated vancomycin minimum inhibitory concentrations within the susceptible range
Authors Phillips C, Wells N, Martinello M, Smith S, Woodman R, Gordon D
Received 7 March 2016
Accepted for publication 31 March 2016
Published 31 May 2016 Volume 2016:9 Pages 87—92
Checked for plagiarism Yes
Review by Single-blind
Peer reviewers approved by Dr Lucy Goodman
Peer reviewer comments 2
Editor who approved publication: Professor Suresh Antony
Cameron J Phillips,1–3 Nicholas A Wells,4 Marianne Martinello,4 Simon Smith,4 Richard J Woodman,5 David L Gordon2,4
1SA Pharmacy, Flinders Medical Centre, Bedford Park, SA, Australia; 2Department of Microbiology and Infectious Diseases, School of Medicine, Flinders University, Adelaide, SA, Australia; 3School of Pharmacy and Medical Sciences, University of South Australia, Adelaide, SA, Australia; 4SA Pathology, Microbiology and Infectious Diseases, Flinders Medical Centre, Bedford Park, SA, Australia; 5Flinders Centre for Epidemiology and Biostatistics, School of Medicine, Flinders University, Adelaide, SA, Australia
Background: Determination of vancomycin minimum inhibitory concentration (MIC) can influence the agent used to treat methicillin-resistant Staphylococcus aureus (MRSA) infection. We studied diagnostic accuracy using E-test and VITEK® 2 against a gold standard broth microdilution (BMD) methodology, the correlation between methods, and associations between vancomycin MIC and MRSA phenotype from clinical isolates.
Methods: MRSA isolates were obtained from April 2012 to December 2013. Vancomycin MIC values were determined prospectively on all isolates by gradient diffusion E-test and automated VITEK® 2 . The Clinical and Laboratory Standards Institute reference BMD method was performed retrospectively on thawed frozen isolates. Diagnostic accuracy for detecting less susceptible strains was calculated at each MIC cutoff point for E-Test and VITEK® 2 using BMD ≥1 µg/mL as a standard. The correlation between methods was assessed using Spearman’s rho (ρ). The association between MRSA phenotype and MIC for the three methods was assessed using Fisher’s exact test.
Results: Of 148 MRSA isolates, all except one (E-test =3 µg/mL) were susceptible to vancomycin (MIC of ≤2 µg/mL) irrespective of methodology. MICs were ≥1.0 µg/mL for 9.5% of BMD, 50.0% for VITEK® 2 , and 27.7% for E-test. Spearman’s ρ showed weak correlations between methods: 0.29 E-test vs VITEK® 2 (P=0.003), 0.27 E-test vs BMD (P=0.001), and 0.31 VITEK® 2 vs BMD (P=0.002). The optimal cutoff points for detecting BMD-defined less susceptible strains were ≥1.0 µg/mL for E-test and VITEK® 2 . E-test sensitivity at this cutoff point was 0.85 and specificity 0.29, while VITEK® 2 sensitivity and specificity were 0.62 and 0.51, respectively. Multiresistant MRSA strains tended to have higher MIC values compared to nonmultiresistant MRSA or epidemic MRSA 15 phenotypes by E-test (Fisher’s exact P<0.001) and VITEK® 2 (Fisher’s exact P<0.001).
Conclusion: Overall diagnostic accuracy and correlations between MIC methods used in routine diagnostic laboratories and the gold standard BMD showed limited overall agreement. This study helps optimize guidance on the effective use of vancomycin.
Keywords: MIC, MRSA, sensitivity, specificity, susceptibility, vancomycin
A letter to the Editor has been recieved and published for this article.
Other article by this author:
Pharmacist-led implementation of a vancomycin guideline across medical and surgical units: impact on clinical behavior and therapeutic drug monitoring outcomes
Phillips CJ, Gordon DL
Published Date: 21 October 2015