Intraocular pressure readings obtained through soft contact lenses using four types of tonometer
Authors Takenaka J, Kunihara E, Rimayanti U, Tanaka J, Kaneko M, Kiuchi Y
Received 19 March 2015
Accepted for publication 25 May 2015
Published 5 October 2015 Volume 2015:9 Pages 1875—1881
Checked for plagiarism Yes
Review by Single anonymous peer review
Peer reviewer comments 4
Editor who approved publication: Dr Scott Fraser
Joji Takenaka,1 Eriko Kunihara,1 Ulfah Rimayanti,2 Junko Tanaka,3 Makoto Kaneko,4 Yoshiaki Kiuchi1
1Department of Ophthalmology and Visual Science, Graduate School of Biomedical Sciences, Hiroshima University, Hiroshima, Japan; 2Faculty of Medicine and Health Sciences, UIN Alauddin Makassar, South Sulawesi, Indonesia; 3Department of Epidemiology, Infectious Disease Control and Prevention, Graduate School of Biomedical Sciences, Hiroshima University, Hiroshima, Japan; 4Graduate School of Engineering, Osaka University, Osaka, Japan
Purpose: To compare the reliability and accuracy of intraocular pressure (IOP) measured while wearing soft contact lenses (SCLs) using a non-contact tonometer (NCT), Goldmann applanation tonometer (GAT), iCare rebound tonometer (RBT) and the Tono-Pen XL.
Methods: Twenty-six healthy subjects were examined. The IOP was measured using NCT, GAT, RBT, and the Tono-Pen XL, while the subjects wore SCLs -5.00 D, -0.50 D and +5.00 D. Bland–Altman plots and a regression analysis were used to compare the IOPs obtained with those instruments and the IOPs of the naked eyes measured using GAT (the standard IOPs in this study).
Results: The IOPs obtained by the Tono-Pen XL while the subjects were wearing -5.00 D, -0.50 D, and +5.00 D SCLs were significantly higher than those of the naked eyes obtained using GAT. RBT showed that the IOPs were similar to the GAT standard IOPs under all conditions. The IOPs measured with NCT and GAT while the subjects were wearing -5.00 D and -0.50 D SCLs were similar to the GAT standard IOPs. The IOPs obtained with RBT and NCT while the subjects were wearing -5.00 D and -0.50 D SCLs exhibited a good correlation with the standard IOPs.
Conclusion: The NCT and RBT are best when measuring IOP through hydrogel SCLs.
Keywords: soft contact lens, intraocular pressure, rebound tonometer, non-contact tonometer
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]