Effect of pupillary dilation on Haigis formula-calculated intraocular lens power measurement by using optical biometry
Authors Khambhiphant B, Sasiwilasagorn S, Chatbunchachai N, Pongpirul K
Received 4 April 2016
Accepted for publication 8 June 2016
Published 27 July 2016 Volume 2016:10 Pages 1405—1410
Checked for plagiarism Yes
Review by Single-blind
Peer reviewers approved by Dr Colin Mak
Peer reviewer comments 4
Editor who approved publication: Dr Scott Fraser
Bharkbhum Khambhiphant,1 Suganlaya Sasiwilasagorn,2 Nattida Chatbunchachai,3 Krit Pongpirul2,4
1Department of Ophthalmology, King Chulalongkorn Memorial Hospital, Thai Red Cross Society, 2Department of Ophthalmology, Faculty of Medicine, Chulalongkorn University, Bangkok, 3Department of Ophthalmology, Samut Prakan Hospital, Samut Prakan, Thailand; 4Department of International Health, Johns Hopkins Bloomberg School of Public Health, Baltimore, MD, USA
Purpose: The purpose of this study was to evaluate the effect of pupillary dilation on the Haigis formula-calculated intraocular lens (IOL) power and ocular biometry measurements by using IOLMaster®.
Methods: A prospective study was performed for biometry measurements of 373 eyes of 192 healthy subjects using the IOLMaster at the outpatient department of King Chulalongkorn Memorial Hospital from February 2013 to July 2013. The axial length (AL), anterior chamber depth (ACD), keratometry (K), and IOL power were measured before and after 1% tropicamide eye drop instillation. The Haigis formula was used in the IOL power calculation with the predicted target to emmetropia. Each parameter was compared by a paired t-test prior to and after pupillary dilation. Bland–Altman plots were also used to determine the agreement between each parameter.
Results: The mean age of the subjects was 53.74±14.41 years (range 18–93 years). No differences in AL (P=0.03), steepest K (P=0.42), and flattest K (P=0.41) were obtained from the IOLMaster after pupillary dilation. However, ACD and IOL power were significantly different postdilation (P<0.01 and P<0.01, respectively). In ACD and IOL power measurements, the concordance rates were 93.03% and 97.05% within 95% limits of agreement (-0.48 to 0.26 mm and -1.09 to 0.88 D, respectively) in the Bland–Altman plots.
Conclusion: Biometry measurements in the cycloplegic stage should be considered in the IOL formulas that use parameters other than AL and K.
Keywords: Haigis formula, intraocular lens, IOLMaster, optical biometry, pupillary dilation
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]