Comparison of the precision of the Topcon SP-3000P specular microscope and an ultrasound pachymeter
Turki M Almubrad, Uchechukwu L Osuagwu, Ibrahim AlAbbadi, Kelechi C Ogbuehi
Cornea Research Chair, Department of Optometry, College of Applied Medical Sciences, King Saud University, Riyadh, Saudi Arabia
Aim: To compare the precision of the Topcon SP-3000P noncontact specular microscope (NCSM) and the DGH 500 ultrasound pachymeter (USP).
Methods: Triplicate measurements of central corneal thickness (CCT) for 100 eyes were taken with an NCSM and a USP in 2 visits separated by 1 week. Repeatability was assessed by computing the differences between all 3 readings from each subject. Coefficients of repeatability and reproducibility were computed.
Results: Mean CCT as measured by each instrument were: 518.53 ± 34.96 µm (range 417.33–592.67) and 516.94 ± 33.60 µm (range 431.67–582.67) for sessions 1 and 2 respectively, with the NCSM; 546.69 ± 36.62 µm (range 457.33–617.00) and 549.78 ± 35.26 µm (range 454.00–618.67) for sessions 1 and 2 respectively, with the USP. The ultrasound CCT measurements were consistently higher than those obtained with the NCSM in both sessions 28.17 ± 19.20 µm (mean ± SD, session 1) and 32.81 ± 14.04 (mean ± SD, session 2). The repeatability coefficient for the NCSM was better in both sessions than those for USP (±10 µm vs ± 12 µm in session 1 and ±8 µm vs ±10 µm in session 2). The reproducibility coefficient with the NCSM was half that with the USP (±21 µm vs ±41 µm).
Conclusion: The SP-3000P NCSM is a more precise and reproducible instrument for measurement of CCT than the USP, but both instruments are reliable, useful instruments for measuring CCT.
Keywords: cornea, Topcon SP-3000P, ultrasound pachymetry, repeatability coefficient, reproducibility coefficient
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]