Comparison of a new PillCam™ SB2 video capsule versus the standard PillCam™ SB for detection of small bowel disease
Yoav C Metzger1, Samuel N Adler2, Ariella Bar-Gil Shitrit3, Binyamin Koslowsky4, Ingvar Bjarnason5
1School of Medicine, Hebrew University of Jerusalem, Israel; 2Department of Gastroenterology, Bikur Holim Hospital, Jerusalem, Israel; 3Department of Gastroenterology, Shaare Zedek Medical Center, Jerusalem, Israel; 4Department of Medicine, Hadassah Medical Organization, Jerusalem, Israel; 5Department of Gastroenterology, King’s College Hospital, London, UK
Background: Capsule endoscopy is the procedure of choice for many suspected small bowel diseases. PillCam™ small bowel (SB), has a high diagnostic yield. No improvements have been made on its optics since its introduction in 2002.
Aim: To compare the diagnostic yield and quality of visualization of PillCam™ SB with that of a new small bowel endoscope (PillCam™ SB2) which has a larger angle of view and upgraded visual attributes.
Methods: Twenty patients were randomized to undergo capsule enteroscopy with PillCam™ SB and PillCam™ SB2 a week apart.
Results: Both procedures had a comparable diagnostic yield. However images from PillCam™ SB2 were found to be significantly better in regards of homogenous light exposure (P < 0.001), sharpness and resolution (P < 0.001), depth of view (P < 0.005), and overall impression (P < 0.0001).
Conclusion: PillCam™ SB2 appears to be superior to the original PillCam™ SB in regard to quality of images obtained.
Keywords: small bowel, capsule endoscopy, image quality, resolution
This work is published by Dove Medical Press Limited, and licensed under Creative Commons Attribution - Non Commercial (unported, v3.0) License. The full terms of the License are available at http://creativecommons.org/licenses/by-nc/3.0/. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. Permissions beyond the scope of the License are administered by Dove Medical Press Limited. Information on how to request permission may be found at: http://www.dovepress.com/permissions.php
Readers of this article also read:
Yao J, Li Y, Sun X, Dahmani FZ, Liu H, Zhou J
Published Date: 10 July 2014
Tan JCH, Heng WJ
Published Date: 25 February 2013
Bäumler M, Bigorre M, Faure J-M
Published Date: 3 December 2012
Consideration of generated beam angles increases the accuracy of ultrasonic displacement measurements
Sumi C, Takanashi Y, Ichimaru K
Published Date: 9 March 2012
Subtle cytotoxicity and genotoxicity differences in superparamagnetic iron oxide nanoparticles coated with various functional groups
Hong SC, Lee JH, Lee J, Kim HY, Park JY, Cho J, Lee J, Han DW
Published Date: 7 December 2011
Forssell-Aronsson E, Schüler E, Ahlman H
Published Date: 11 May 2011
Polyurethane/fluor-hydroxyapatite nanocomposite scaffolds for bone tissue engineering. Part I: morphological, physical, and mechanical characterization
Azadeh Asefnejad, Aliasghar Behnamghader, Mohammad Taghi Khorasani, et al
Published Date: 6 January 2011
Stephanie L Hines
Published Date: 28 October 2010
Brian Mucci, Robert Shaw
Published Date: 9 June 2010
Michael Harzheim, Claudia Gemmel, Bernd Sommer, Ulf Kallweit, Dieter Pöhlau
Published Date: 24 November 2008