Agreement between retinal images obtained via smartphones and images obtained with retinal cameras or fundoscopic exams – systematic review and meta-analysis
Received 31 July 2018
Accepted for publication 10 October 2018
Published 11 December 2018 Volume 2018:12 Pages 2581—2589
Checked for plagiarism Yes
Review by Single-blind
Peer reviewers approved by Dr Amy Norman
Peer reviewer comments 2
Editor who approved publication: Dr Scott Fraser
Manuel AP Vilela,1,2 Felipe M Valença,1 Pedro KM Barreto,1 Carlos EV Amaral,1 Lúcia C Pellanda1,2
1Federal University of Health Sciences of Porto Alegre, Porto Alegre, Rio Grande do Sul, Brazil; 2Institute of Cardiology, Cardiology University Foundation, Porto Alegre, Rio Grande do Sul, Brazil
Background: Smartphone fundoscopy is a new option for visualizing the ocular fundus but must be validated before being included in population-based examinations. Our aim was to evaluate the quality of fundoscopic images obtained via smartphone and to compare their agreement with retinal camera images or clinical examination.
Methods: The database for this study included all observational studies with smartphone fundoscopy that have comparative analyses with the gold standard methods.
Results: Out of 121 potentially relevant studies, nine were included in this analysis, compris- ing a total of 4,219 eyes. Mean age was 56.6 years (SD±8.5). Combined kappa (κ) agreement statistics were equal to 77.77% (95% CI: 70.34%, 83.70%). No heterogeneity was measured by random effects (I2=zero).
Conclusion: Fundoscopic images obtained by using smartphones have substantial agreement with gold standards for clinical or photographic exams.
Keywords: smartphones, fundoscopy, ophthalmoscopy
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]