Visual field examination method using virtual reality glasses compared with the Humphrey perimeter
Received 27 December 2016
Accepted for publication 13 April 2017
Published 7 August 2017 Volume 2017:11 Pages 1431—1443
Checked for plagiarism Yes
Review by Single anonymous peer review
Peer reviewer comments 2
Editor who approved publication: Dr Scott Fraser
Stylianos Tsapakis, Dimitrios Papaconstantinou, Andreas Diagourtas, Konstantinos Droutsas, Konstantinos Andreanos, Marilita M Moschos, Dimitrios Brouzas
1st Department of Ophthalmology, National and Kapodistrian University of Athens, Athens, Greece
Purpose: To present a visual field examination method using virtual reality glasses and evaluate the reliability of the method by comparing the results with those of the Humphrey perimeter.
Materials and methods: Virtual reality glasses, a smartphone with a 6 inch display, and software that implements a fast-threshold 3 dB step staircase algorithm for the central 24° of visual field (52 points) were used to test 20 eyes of 10 patients, who were tested in a random and consecutive order as they appeared in our glaucoma department. The results were compared with those obtained from the same patients using the Humphrey perimeter.
Results: High correlation coefficient (r=0.808, P<0.0001) was found between the virtual reality visual field test and the Humphrey perimeter visual field.
Conclusion: Visual field examination results using virtual reality glasses have a high correlation with the Humphrey perimeter allowing the method to be suitable for probable clinical use.
Keywords: visual fields, virtual reality glasses, perimetry, visual fields software, smartphone
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]