Validation of four automatic devices for self-measurement of blood pressure according to the international protocol of the European Society of Hypertension
Jirar Topouchian1, Davide Agnoletti1, Jacques Blacher1, Ahmed Youssef1, Isabel Ibanez2,3, Jose Khabouth2, Salwa Khawaja2, Layale Beaino2, Roland Asmar1–3
1Centre de Diagnostic, Hôpital Hôtel-Dieu, Paris, France; 2Hôpital Libanais and Faculté Libanaise de Médecine, Beirut, Lebanon; 3Foundation-Medical Research Institutes, Geneva, Switzerland
Background: Four oscillometric devices for self-measurement of blood pressure (SBPM) were evaluated according to the European Society of Hypertension (ESH) international protocol and its 2010 revision in four separate studies. The Omron® M2, Omron M3, and Omron M6 measure blood pressure (BP) at the brachial level, while the Omron R2 measures BP at the wrist level.
Methods: The international protocol requires a total number of 33 subjects in which the validation is performed. The Omron M2 and Omron R2 were validated in 2009 according to the ESH international protocol, while the Omron M3 and Omron M6 were validated in 2010–2011 according to the 2010 ESH international protocol revision. The protocol procedures were followed precisely.
Results: All four tested devices passed the validation process. The mean differences between the device and mercury readings were 2.7 ± 5.0 and –1.4 ± 3.2 mmHg for systolic and diastolic BP, respectively, using the Omron M2 device, and 1.7 ± 3.2 and –0.9 ± 2.6 mmHg using the Omron M3, 1.6 ± 2.9 and -0.9 ± 2.5 mmHg using the Omron M6, and –1.1 ± 4.8 and –0.9 ± 4.3 mmHg using the Omron R2.
Conclusion: Readings from the Omron M2, Omron M3, Omron M6, and Omron R2, differing by less than 5, 10, and 15 mmHg, fulfill the ESH international protocol and its 2010 revision requirements. Therefore, each of these four devices can be used by patients for SBPM.
Keywords: Omron R2, M2, M3, M6, blood pressure measurement, validation, international protocol, European Society of Hypertension
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]