Validation of two automatic devices, Omron HEM-6232T and HEM-6181, for self-measurement of blood pressure at the wrist according to the ANSI/AAMI/ISO 81060-2:2013 protocol and the European Society of Hypertension International Protocol revision 2010
Authors Saito K, Hishiki Y, Takahashi H
Received 19 September 2018
Accepted for publication 16 January 2019
Published 6 March 2019 Volume 2019:15 Pages 47—55
Checked for plagiarism Yes
Review by Single-blind
Peer reviewer comments 3
Editor who approved publication: Prof. Dr. Bojan Jelaković
Kanako Saito,1 Yukiko Hishiki,1 Hakuo Takahashi2
1Department of Technology Development, Omron Healthcare Co, Ltd., Mukou City, Kyoto, Japan; 2Department of Cardiology, Biwako Central Hospital, Otsu City, Shiga, Japan
Objective: The performance of Omron HEM-6232T and Omron HEM-6181 for monitoring blood pressure (BP) at the wrist was validated in accordance with the ANSI/AAMI/ISO 81060-2:2013 protocol (ANSI/AAMI/ISO) and the European Society of Hypertension International Protocol revision 2010 (ESH IP2).
Methods: Three trained medical technologists validated the performance of these devices by comparing data obtained from these devices with those obtained using a standard mercury sphygmomanometer.
Results: The mean differences between the devices and mercury readings for SBP and DBP were as follows: HEM-6232T, –0.4±6.7 mmHg and 1.6±5.4 mmHg, respectively; HEM-6181, –0.7±6.2 mmHg and –0.7±5.2 mmHg, respectively, satisfying the ANSI/AAMI/ISO protocol. The mean device–observer measurement difference was –0.9±5.7 mm Hg and 0.2±4.6 mm Hg for SBP and 0.5±4.9 mm Hg and 1.4±3.5 mm Hg for DBP, for HEM-6232T and HEM-6181, respectively, satisfying part 1 of the ESH-IP2. All differences for SBP and DBP in both devices satisfied part 2 of the ESH-IP2. The number of absolute differences in the values obtained using the devices and those measured by the observers fulfilled the requirements of the ANSI/AAMI/ISO and the ESH IP2.
Conclusion: The Omron HEM-6232T and HEM-6181 devices met all the requirements of the ANSI/AAMI/ISO and the ESH IP2.
Keywords: blood pressure measurement, device, guideline, international protocol, validation
Corrigendum for this paper has been published
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]