Peak inspiratory flow rate measurement by using In-Check DIAL for the different inhaler devices in elderly with obstructive airway diseases
Authors Kawamatawong T, Khiawwan S, Pornsuriyasak P
Received 14 November 2016
Accepted for publication 19 January 2017
Published 23 February 2017 Volume 2017:10 Pages 17—21
Checked for plagiarism Yes
Review by Single-blind
Peer reviewers approved by Dr Amy Norman
Peer reviewer comments 2
Editor who approved publication: Dr Amrita Dosanjh
Theerasuk Kawamatawong, Supattra Khiawwan, Prapaporn Pornsuriyasak
Division of Pulmonary and Critical Care Medicine, Department of Medicine, Ramathibodi Hospital, Mahidol University, Bangkok, Thailand
Background: Inhaler device technique is a common cause of treatment failure in patients with asthma and chronic obstructive pulmonary disease. Dry powder inhaler (DPI) requires optimal peak inspiratory flow rate (PIFR) for drug delivery. Low PIFR generation is common in the elderly. Patient lung function and intrinsic inhaler resistance are factors for determining generated PIFR and drug delivery from DPI.
Objectives: We aimed to identify the PIFR of the older (aged >60 years) and the younger (aged ≤60 years) patients with obstructive airway diseases for the different inhaler devices (Turbuhaler® and Accuhaler).
Patients and methods: A cross-sectional study was conducted from January to December 2014. Patients with obstructive airway diseases were recruited. Spirometry was performed. PIFR was measured by using an In-Check DIAL device. Individual PIFR values for each inhaler device were obtained for three consecutive measurements and then averaged.
Results: A total of 139 patients diagnosed with obstructive lung diseases (asthma, n = 109; chronic obstructive pulmonary disease, n = 30) were recruited. Of these, 71 patients (51%) were >60 years. The PIFR generated by the patients who were ≤60 years for nonresistance mode was not different from that generated by those aged >60 years (115.0 ± 15.2 L/min vs 115.4 ± 13.3 L/ min, p = 0.86). Regarding the DPI, PIFR generated from the older group was significantly lower than that generated from the younger group for Turbuhaler (72.5 ± 18.8 L/min vs 82.4 ± 21.1 L/min, p = 0.01), but the PIFR generated was not significantly different between the older and the younger groups for the Accuhaler (93.8 ± 22.9 L/min vs 99.4 ± 24.2 L/min, p = 0.86). The low peak expiratory flow rate and PIFR from spirometry were associated with the suboptimal PIFR measured by using In-Check DIAL.
Discussion: Optimal PIFR is critical for DPI use in the elderly; appropriate DPI selection is essential for management. In-Check DIAL may be useful for detecting inhaler device problem among the elderly.
Conclusion: Lower PIFR generated from Turbuhaler was noted in patients with airway diseases who were older than 60 years, when compared to the younger patients.
Keywords: elderly, obstructive airway diseases, peak inspiratory flow rate, dry powder inhaler
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]