Therapeutics and Clinical Risk Management
Open access peer-reviewed scientific and medical journals.
Dove Medical Press is now a member of the Open Access Initiative
An Author's Guide
A guide to help authors get their paper published.
Support Open Access and Dove Press
Promotional Article Monitoring - further details
Favored Author Program
Real benefits for authors, including fast-track processing of papers.
An improved delivery system for bladder irrigation
(8534) Total Article Views
Authors: Mohammad K Moslemi, Mojtaba Rajaei
Published Date September 2010
Volume 2010:6 Pages 459 - 462
Mohammad K Moslemi1, Mojtaba Rajaei2
1Department of Urology, 2Kamkar Hospital, School of Medicine, Qom University of Medical Sciences, Qom, Iran
Introduction: Occasionally, urologists may see patients requiring temporary bladder irrigation at hospitals without stocks of specialist irrigation apparatus. One option is to transfer the patient to a urology ward, but often there are outstanding medical issues that require continued specialist input. Here, we describe an improved system for delivering temporary bladder irrigation by utilizing readily available components and the novel modification of a sphygmomanometer blub. This option is good for bladder irrigation in patients with moderate or severe gross hematuria due to various causes.
Materials and methods: In this prospective study from March 2007 to April 2009, we used our new system in eligible cases. In this system, an irrigant bag with 1 L of normal saline was suspended 80 cm above the indwelled 3-way Foley catheter, and its drainage tube was inserted into the irrigant port of the catheter. To increase the flow rate of the irrigant system, we inserted a traditional sphygmomanometer bulb at the top of the irrigant bag. This closed system was used for continuous bladder irrigation (CBI) in patients who underwent open prostatectomy, transurethral resection of the prostate (TURP), or transurethral resection of the bladder (TURB). This high-pressure system is also used for irrigation during cystourethroscopy, internal urethrotomy, and transurethral lithotripsy. Our 831 eligible cases were divided into two groups: group 1 were endourologic cases and group 2 were open prostatectomy, TURP, and TURB cases. The maximum and average flow rates were evaluated. The efficacy of our new system was compared prospectively with the previous traditional system used in 545 cases.
Results: In group 1, we had clear vision at the time of endourologic procedures. The success rate of this system was 99.5%. In group 2, the incidence of clot retention decreased two-fold in comparison to traditional gravity-dependent bladder flow system. These changes were statistically significant (P = 0.001). We did not observe any adverse effects such as bladder perforation due to our high-pressure, high-flow system.
Conclusion: A pressurized irrigant system has better visualization during endourologic procedures, and prevents clot formation after open prostatectomy, TURP, and TURB without any adverse effects.
Keywords: cystoscopy, drainage, pressure, open prostatectomy, sphygmomanometer, transurethral lithotripsy, transurethral resection, prostate, bladder
Cannotea Citeulike Del.icio.us Facebook LinkedIn Twitter
Other articles by Professor Mohammad Kazem Moslemi
Readers of this article also read:
- American Acne and Rosacea Society
The American Acne and Rosacea Society (AARS) is a 501(c)(6) non-profit organization dedicated to elevating the understanding and treatment of acne and rosacea.
- Enzalutamide: an evidence-based review of its use in the treatment of prostate cancer
- The benefits and risks of testosterone replacement therapy: a review
- Tumor necrosis factor-alpha inhibitor treatment for sarcoidosis
- Tenofovir-associated bone density loss