Validation of the custo screen 400 ambulatory blood pressure-monitoring device according to the European Society of Hypertension International Protocol revision 2010
Authors Bramlage P, Deutsch C, Krüger R, Wolf A, Müller P, Zwingers T, Beime B, Mengden T
Received 5 March 2014
Accepted for publication 4 April 2014
Published 13 May 2014 Volume 2014:10 Pages 303—309
Checked for plagiarism Yes
Review by Single-blind
Peer reviewer comments 3
Peter Bramlage,1 Cornelia Deutsch,1 Ralf Krüger,1 Andreas Wolf,2 Peter Müller,2 Thomas Zwingers,1,4 Beate Beime,1 Thomas Mengden3
1Institut für Pharmakologie und Präventive Medizin, Cloppenburg, 2Müller and Sebastiani, Ottobrunn, 3Kerckhoff-Klinik, Bad Nauheim, 4Estimate, Augsburg, Germany
Objective: The aim of the present study was to validate the custo screen 400 ambulatory blood pressure-monitoring (ABPM) device according to the 2010 International Protocol revision of the European Society of Hypertension (ESH-IP). The device can be used for ABPM for up to 72 hours.
Materials and methods: Systolic and diastolic blood pressure (SBP and DBP, respectively) were sequentially measured in 33 adult subjects (13 males and 20 females) and compared with a standard mercury sphygmomanometer (two observers). A total of 99 comparison pairs were obtained.
Results: The custo screen 400 met the requirements of parts 1 and 2 of the ESH-IP revision 2010. The mean difference between the device and reference sphygmomanometer readings was −0.5±4.5 mmHg for SBP and −0.1±3.3 mmHg for DBP. All but one measurement were within the absolute difference of 10 mmHg between the device and the observers for SBP and DBP. The number of absolute differences between the device and the observers within a range of 5 mmHg was 84 of 99 readings for SBP, and 93 of 99 readings for DBP.
Conclusion: The custo screen 400 ABPM device met the requirements of the 2010 ESH-IP revision, and hence can be recommended for ABPM in adults. To our knowledge, the custo screen 400 is the first device to pass the revised ESH-IP 2010.
Keywords: validation, ambulatory blood pressure monitoring, ESH
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]