Refractive cylinder outcomes after calculating toric intraocular lens cylinder power using total corneal refractive power
Authors Davison J, Potvin R
Received 16 May 2015
Accepted for publication 9 July 2015
Published 19 August 2015 Volume 2015:9 Pages 1511—1517
Checked for plagiarism Yes
Review by Single anonymous peer review
Peer reviewer comments 4
Editor who approved publication: Dr Scott Fraser
James A Davison,1 Richard Potvin2
1Wolfe Eye Clinic, Marshalltown, IA, USA; 2Science in Vision, Akron, NY, USA
Purpose: To determine whether the total corneal refractive power (TCRP) value, which is based on measurement of both anterior and posterior corneal astigmatism, is effective for toric intraocular lens (IOL) calculation with AcrySof® Toric IOLs
Patients and methods: A consecutive series of cataract surgery cases with AcrySof toric IOL implantation was studied retrospectively. The IOLMaster® was used for calculation of IOL sphere, the Pentacam® TCRP 3.0 mm apex/ring value was used as the keratometry input to the AcrySof Toric IOL Calculator and the VERION™ Digital Marker for surgical orientation. The keratometry readings from the VERION reference unit were recorded but not used in the actual calculation. Vector differences between expected and actual residual refractive cylinder were calculated and compared to simulated vector errors using the collected VERION keratometry data.
Results: In total, 83 eyes of 56 patients were analyzed. Residual refractive cylinder was 0.25 D or lower in 58% of eyes and 0.5 D or lower in 80% of eyes. The TCRP-based calculation resulted in a statistically significantly lower vector error (P<0.01) and significantly more eyes with a vector error ≤0.5 D relative to the VERION-based calculation (P=0.02). The TCRP and VERION keratometry readings suggested a different IOL toric power in 53/83 eyes. In these 53 eyes the TCRP vector error was lower in 28 cases, the VERION error was lower in five cases, and the error was equal in 20 cases. When the anterior cornea had with-the-rule astigmatism, the VERION was more likely to suggest a higher toric power and when the anterior cornea had against-the-rule astigmatism, the VERION was less likely to suggest a higher toric power.
Conclusion: Using the TCRP keratometry measurement in the AcrySof toric calculator may improve overall postoperative refractive results. Consideration of measured posterior corneal astigmatism, rather than a population-averaged value, appears advantageous.
Keywords: toric IOL, posterior corneal astigmatism, astigmatism, keratometry, biometry, TCRP
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]