Optimization of the Oertli CataRhex 3® phacoemulsification machine
Received 4 December 2018
Accepted for publication 20 February 2019
Published 16 April 2019 Volume 2019:13 Pages 633—639
Checked for plagiarism Yes
Review by Single anonymous peer review
Peer reviewer comments 3
Editor who approved publication: Dr Scott Fraser
Lance A Stutz,1 Joshua B Heczko,2 Brian A Bird,3 Rhett S Thomson,2,4 Ashlie A Bernhisel,2 William R Barlow,2 Brian Zaugg,2 Randall J Olson,2 Jeff H Pettey2
1University of Texas Southwestern Medical School, University of Texas Southwestern Medical Center, Dallas, TX 75390, USA; 2Department of Ophthalmology and Visual Sciences, John A Moran Eye Center, University of Utah, Salt Lake City, UT 84132, USA; 3University of Nevada Reno, School of Medicine, Reno, NV 89557, USA; 4University of Utah, School of Medicine, Salt Lake City, UT 84132, USA
Purpose: The aim of this study was to determine optimal bottle height, vacuum, aspiration rate, and power settings of the Oertli CataRhex 3® phacoemulsification machine.
Methods: Porcine lens nuclei were hardened with formalin and cut into 2.0 mm cubes. Lens cubes were emulsified using the easyTip® 2.2 mm at 30°. Fragment removal time (efficiency) and fragment bounces off the tip (chatter) were measured. Settings tested included bottle height of 60, 80, 100 and 120 cm; aspiration rate of 40, 45, and 50 mL/min; vacuum of 400, 500, and 600 mmHg; and power of 50, 60, 70, 80, 90, and 100%.
Results: Efficiency and chatter increased in a linear fashion with increasing vacuum to 600 mmHg (P=0.017, P=0.046, respectively). The most efficient aspiration rate was 50 mL/min, although this finding lacked statistical significance (P=0.66). Increasing power increased efficiency up to 80% without increasing chatter (P=0.042, P=0.71, respectively). Compared to all other power settings, chatter was increased at 100% (P=0.014).
Conclusion: The most efficient machine settings were vacuum at 600 mmHg, aspiration rate at 50 mL/min, and power at 80%.
Keywords: cataract, blindness, underserved countries, vacuum, aspiration rate, power
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]