Measurement of central corneal thickness by ultrasonic pachymeter and oculus pentacam in patients with well-controlled glaucoma: hospital-based comparative study
Authors Rashid R, Farhood Q
Received 14 September 2015
Accepted for publication 3 December 2015
Published 1 March 2016 Volume 2016:10 Pages 359—364
Checked for plagiarism Yes
Review by Single-blind
Peer reviewer comments 2
Editor who approved publication: Dr Scott Fraser
Riyam Faihan Rashid, Qasim K Farhood
Department of Ophthalmology, College of Medicine, University of Babylon, Babylon, Iraq
Background: The measurement of central corneal thickness (CCT) plays an important role in the diagnosis and treatment of glaucoma and many corneal diseases.
Objective of the study: To compare the measurement of CCT by ultrasonic pachymeter with that measured by oculus pentacam in both normal subjects and patients with well-controlled glaucoma.
Patients and methods: In 173 eyes of both controls and patients with open-angle glaucoma (normal intraocular pressure) attending Ibn Al Haitham Teaching Eye Hospital, CCT was measured by oculus pentacam and then by ultrasound pachymeter at the same morning by the same technicians.
Results: The results showed no significant difference in CCT readings measured by the two devices in both glaucoma and control groups (glaucoma group: 547.29±49.32 µm with pentacam vs 547.66±45.24 µm with ultrasound pachymeter; control group: 551.02±36.28 µm with pentacam vs 541.25±34.96 µm with ultrasound pachymeter). P-values were >0.05 in both groups (statistically not significant).
Conclusion and recommendation: Ultrasonic pachymeter and oculus pentacam can be used interchangeably in measuring CCT, and we recommend a nontouch method (in this study, pentacam Scheimpflug camera) for measuring CCT during assessment of patients with glaucoma or any ocular disease or surgery.
Keywords: CCT, open-angle glaucoma, ultrasound pachymeter, Scheimpflug camera
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]