Comparison of rechargeable versus battery-operated insulin pumps: temperature fluctuations
Authors Vereshchetin P, McCann Jr TW, Ojha N, Venugopalan R, Levy BL
Received 7 July 2016
Accepted for publication 17 September 2016
Published 14 October 2016 Volume 2016:9 Pages 371—376
Checked for plagiarism Yes
Review by Single-blind
Peer reviewers approved by Dr Akshita Wason
Peer reviewer comments 2
Editor who approved publication: Dr Scott Fraser
Paul Vereshchetin, Thomas W McCann Jr, Navdeep Ojha, Ramakrishna Venugopalan, Brian L Levy
Johnson & Johnson Diabetes Care Companies, Chesterbrook, PA, USA
Abstract: The role of continuous subcutaneous insulin infusion (insulin pumps) has become increasingly important in diabetes management, and many different types of these systems are currently available. This exploratory study focused on the reported heating issues that lithium-ion battery-powered pumps may have during charging compared with battery-operated pumps. It was found that pump temperature increased by 6.4°C during a long charging cycle of a lithium-ion battery-operated pump under ambient temperatures. In an environmental-chamber kept at 35°C, the pump temperature increased by 4.4°C, which indicates that the pump temperature was above that of the recommended safety limit for insulin storage of 37°C. When designing new pumps, and when using currently available rechargeable pumps in warmer climates, the implications of these temperature increases should be taken into consideration. Future studies should also further examine insulin quality after charging.
Keywords: insulin pumps, diabetes mellitus, safety, heating
A Letter to the Editor has been received and published for this article.
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]