Comparison of choroidal thickness measurements between spectral-domain OCT and swept-source OCT in normal and diseased eyes
Authors Zafar S, Siddiqui MAR, Shahzad R
Received 12 July 2016
Accepted for publication 27 September 2016
Published 14 November 2016 Volume 2016:10 Pages 2271—2276
Checked for plagiarism Yes
Review by Single-blind
Peer reviewer comments 3
Editor who approved publication: Dr Scott Fraser
Sidra Zafar,1 MA Rehman Siddiqui,2,3 Rida Shahzad1
1Medical College, Aga Khan University Hospital, 2Department of Ophthalmology, Shahzad Eye Hospital, 3South City Hospital, Karachi, Pakistan
Purpose: Sub-foveal choroidal thickness (SFCT) is affected in many ocular diseases. The aim of this study was to compare SFCT measurements between Topcon 3D 2000 spectral-domain optical coherence tomography (SD-OCT) and Topcon swept-source OCT (SS-OCT), with different laser wavelengths, in normal and diseased populations.
Materials and methods: This was a prospective, cross-sectional, noninterventional study including 27 normal volunteers and 27 participants with retinal disease. OCT scans were performed sequentially and under standardized conditions using both SD-OCT and SS-OCT. The OCT scans were evaluated by two independent graders. Paired t-tests and intraclass correlation coefficients (ICCs) were used to assess the statistically significant difference between SFCT measurements as measured by the two devices.
Results: Mean SFCT measurements for all 54 participants were 264.9±103.1 µm using SD-OCT (range: 47–470 µm) and 278.5±110.5 µm using SS-OCT (range: 56–502 µm), with an inter-device ICC of 0.850. Greater variability was noted in the diseased eyes. Inter-device ICCs were 0.870 (95% CI; 0.760–0.924) and 0.840 (95% CI; 0.654–0.930) for normal and diseased eyes, respectively. However, the difference was not statistically significant (P=0.132).
Conclusion: Both machines reliably measure SFCT. Larger studies are needed to confirm these findings.
Keywords: choroidal imaging, diseased, normal, SD-OCT, SS-OCT
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]