Characterization of outcomes 1 year after endoscopic thermal vapor ablation for patients with heterogeneous emphysema
Authors Herth F, Ernst A, Baker, Egan, Gotfried M, Hopkins, Stanzel, Valipour A, Wagner, Witt, Kesten S, Snell
Received 21 February 2012
Accepted for publication 28 March 2012
Published 18 July 2012 Volume 2012:7 Pages 397—405
Review by Single-blind
Peer reviewer comments 4
Felix JF Herth,1 Armin Ernst,2 Kimberly M Baker,3 Jim J Egan,4 Mark H Gotfried,5 Peter Hopkins,6 Franz Stanzel,7 Arschang Valipour,8 Manfred Wagner,9 Christian Witt,10 Steven Kesten,11 Gregory Snell12
1Pneumology and Critical Care Medicine, Thoraxklinik Heidelberg, Heidelberg, Germany; 2St Elizabeth's Medical Center, Boston, MA, USA; 3University of Iowa, Iowa City, IA, USA; 4Advanced Lung Disease Program, Mater Misericordiae University Hospital, Dublin, Ireland; 5Pulmonary Associates, Phoenix, AZ, USA; 6Lung Transplant Unit, Prince Charles Hospital, Chermside, Australia; 7Zentrum für Pneumologie, Hemer, Germany; 8Ludwig-Boltzmann-Institute for COPD, Otto-Wagner-Hospital, Vienna, Austria; 9Klinikum Nürnberg, Nürnberg, Germany; 10Pneumology, Charité Campus-Mitte, Berlin, Germany; 11Uptake Medical Corp, Tustin, CA, USA; 12Allergy Immunology and Respiratory Medicine, The Alfred Hospital, Melbourne, Australia
Introduction: Endoscopic lung volume reduction has been developed as a therapeutic option for advanced emphysema. Six-month results following treatment with endoscopic thermal vapor ablation (InterVapor™; Uptake Medical, Tustin, CA) were described previously, and here we report observations from the 12-month assessment.
Methods: Two multicenter, international, single-arm trials of InterVapor (unilateral upper lobe treatment) in patients with upper lobe predominant emphysema were conducted. Inclusion criteria: forced expiratory volume in 1 second (FEV1) 15%–45% predicted, residual volume > 150%, total lung capacity > 100%, 6-minute walk distance (6MWD) > 140 m, and diffusing capacity for carbon monoxide > 20% predicted. Efficacy endpoints: spirometry, body plethysmography, lung volumes by high-resolution computed tomography, St George's Respiratory Questionnaire, modified Medical Research Council dyspnea scale, and 6MWD. All adverse events were collected and independently adjudicated.
Results: Forty four patients were treated at a mean (standard deviation) age of 63 (5.6) years, FEV1 0.86 mL (0.25 mL) (n = 22 men and 22 women). Mean (standard deviation) changes from baseline at 12 months were: FEV1 86.2 mL (173.8 mL), St George's Respiratory Questionnaire -11.0 (14.0) units, treated lobar volume from high-resolution computed tomography -751.8 mL (653.9 mL), residual volume -302.8 mL (775.6 mL), 6MWD 18.5 m (63.7 m), and modified Medical Research Council dyspnea scale score -0.83 (0.97) (P < 0.05 for all except 6MWD). Improvements were numerically larger at 6 versus 12 months. GOLD stage III and IV patients had similar outcomes at 6 months; however, improvements relative to baseline were numerically higher in GOLD stage IV patients. Larger improvements were observed in patients with higher heterogeneity. In total, 39 serious adverse events were reported in 23 patients with 10 events in 8 patients between 6 and 12 months.
Conclusion: Unilateral lobar InterVapor treatment of heterogeneous emphysema improved lung function and health outcomes 1 year following treatment. The magnitude of improvement was larger at 6 months compared to 12 months. Improvements relative to baseline continue to be exhibited at 12 months despite the expected disease related decline over time.
Clinical trials: NCT 01041586 and NCT 01102712
Keywords: emphysema, bronchoscopy, lung volume reduction, thermal energy
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]