Biomaterial suture Vicryl Plus reduces wound-related complications
Authors Zhang Q, Zhang C, Fang X, Luo X, Guo J
Received 5 February 2018
Accepted for publication 8 May 2018
Published 13 August 2018 Volume 2018:14 Pages 1417—1421
Checked for plagiarism Yes
Review by Single-blind
Peer reviewer comments 3
Editor who approved publication: Professor Deyun Wang
Qian Zhang, Chao Zhang, Xuanyu Fang, Xu Luo, Jiwei Guo
Southwest Hospital, The Third Military Medical University (Army Medical University), Chongqing, China
Objective: To compare the wound-related complications following craniotomy in patients with brain gliomas undergoing wound closure using either absorbable antibacterial Vicryl Plus suture or traditional braided silk suture.
Materials and methods: Two hundred and forty-five patients undergoing craniotomy for supratentorial gliomas from January 1, 2011 to December 31, 2013 were retrospectively analyzed. We divided the patients into two groups: the Vicryl Plus group (using absorbable antibacterial Vicryl Plus suture) and the control group (using traditional braided silk suture). The clinical data and wound-related complications after surgery between two groups have been summarized.
Results: Patients in the Vicryl Plus suture group had lower incidence of wound-related complications compared with traditional braided silk suture group (P<0.05). No statistically significant difference was found between the two groups in terms of total medical expense.
Conclusion: Incision closure using absorbable antibacterial Vicryl Plus suture had less wound-related complications than using traditional braided silk suture, without increasing medical expense. Vicryl Plus suture could be a better choice for scalp closure in patients undergoing craniotomy surgery.
Keywords: biomaterial, Vicryl Plus, wound-related complications
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]