A comparison of McGrath MAC, Pentax AWS, and Macintosh direct laryngoscopes for nasotracheal intubation: a randomized controlled trial
Received 22 June 2019
Accepted for publication 18 August 2019
Published 18 September 2019 Volume 2019:15 Pages 1121—1128
Checked for plagiarism Yes
Review by Single-blind
Peer reviewers approved by Dr Nicola Ludin
Peer reviewer comments 3
Editor who approved publication: Professor Garry Walsh
Yun Jeong Chae,* Dae Hee Kim,* Eun Jeong Park, Juyeon Oh, In Kyong Yi
Department of Anesthesiology and Pain Medicine, Ajou University School of Medicine, Suwon 16499, South Korea
*These authors contributed equally to this work
Correspondence: In Kyong Yi
Department of Anaesthesiology and Pain Medicine, Ajou University School of Medicine, 164 World Cup-ro, Yeongtong-gu, Suwon 16499, South Korea
Tel +82 219 7522
Fax +82 219 5579
Purpose: Videolaryngoscopy in nasotracheal intubation has been reported to be better than direct laryngoscopy. The most suitable type of videolaryngoscope remains unknown. This study aimed to compare two videolaryngoscopes (McGrath MAC and Pentax AWS) with a Macintosh laryngoscope during nasotracheal intubation.
Methods: Overall, 123 patients older than 18 with normal airways who needed nasotracheal intubation were randomly allocated into three groups: Macintosh (n=41), McGrath (n=41), and Pentax (n=41). Intubation time was the primary outcome and subdivided into three steps: nose to oropharynx, oropharynx to laryngeal inlet, and laryngeal inlet to trachea. Time required, ease of each step, glottic view grade, modified nasal intubation–difficulty score, and subjective difficulty were evaluated.
Results: Intubation time among the three groups was not significantly different (Macintosh 34.6±8.1 seconds, McGrath 35.2±7.9 seconds, Pentax 36.2±9.7 seconds; p=0.727). While the glottal view was better with videolaryngoscopes (I/IIa/IIb/III 36.6%/36.6%/19.5%/7.3% vs 82.9%/9.8%/7.3%/0%, vs 63.4%/29.3%/4.9%/2.4%, p=0.000), modified nasal intubation–difficulty score and subjective difficulty and ease of each step were not significantly different. However, the Pentax took longest for the second step (11.8±6.3 vs 10.3±3.5 vs 15.1±7.6 seconds, p=0.001) but was shortest for the third step (2.9±2.6 vs 4.4±5.6 vs 1.7±0.7 seconds, p=0.001).
Conclusion: The McGrath MAC and Pentax AWS showed no benefits in intubation time or difficulty, despite better glottal views, compared to the Macintosh laryngoscope in nasotracheal intubation. Additionally, videolaryngoscopes had variable performance at different steps of nasotracheal intubation.
Keywords: laryngoscopes, laryngoscopy, intubation, trachea, nose
This work is published and licensed by Dove Medical Press Limited. The full terms of this license are available at https://www.dovepress.com/terms.php and incorporate the Creative Commons Attribution - Non Commercial (unported, v3.0) License. By accessing the work you hereby accept the Terms. Non-commercial uses of the work are permitted without any further permission from Dove Medical Press Limited, provided the work is properly attributed. For permission for commercial use of this work, please see paragraphs 4.2 and 5 of our Terms.Download Article [PDF] View Full Text [HTML][Machine readable]